The CMS60D Handheld Pulse Oximeter is FDA-approved for digital measurements of SpO2 and pulse rate. Users can select a probe type that meets their requirements for obtaining an accurate measurement. The external probe sensors use comfortable silicone materials and are available in adult, pediatric, soft, wrap, and ear clip models. The CMS60D measures activity levels in the body with an accuracy of ±2bpm for pulse rate and ±2% for blood oxygen saturation levels. External probes are plugged directly into the communication cable socket or through the available probe adapter.
The 1.8” color OLED screen displays detailed measurement results using sharp text and images. Measurement results update in real-time as a numeric value, bar graph, and waveform animation. Within the CMS60D’s on-screen menu, users can adjust options for probe usage, measurement limit alerts, the clock function, the pulse rate tone indicator, and select the “Routine Check Mode” or “Continuous Monitoring Mode.” Large buttons located on the front of the monitor operate the device’s measurement functions and menu navigation. The screen’s adjustable brightness level enables comfortable viewing in the presence of dark or low-light conditions.
Data history can be transferred to a computer for detailed reference or analysis by a medical health professional. This blood oxygen meter integrates with the included software to link to a PC for transferring data via the USB transfer cable accessory. A continuous monitoring mode is available in the software for viewing the monitor’s input readings. Users can also record and store tests directly on the machine’s built-in storage.
View Interactive 3D Model
Ear Clip Probe
CMS-60D Product Video
|SpO2 Measuring Range: 0% ~ 100% (the resolution is 1%)
|SpO2 Accuracy: 70% ~ 100%: ±2%, below 70% unspecified
|Pulse Rate Measuring Range: 30bpm ~ 250bpm (the resolution is 1bpm)
|Pulse Rate Accuracy: ±2bpm or ±2% (select larger)
|Battery Requirement: 2x AA
|Oximeter Probe Wavelength: 660nm 880nm
|Dimensions: 110(L) x 60(W) x 23(H) mm
|Weight: ~122g (including AA batteries)
Love it! It is accurate and reads quickly. I bought it for the ear attachment. It works great!
Upgraded my pulse odometer to the CMS60- handheld. Only the best for my clients! It’s streamlined, effective, and accurate.
finger probe works good
Very small size but does everything advertised. Bought a 2nd one for a back up.